DT1 (DT5) card reader
How to use the special card reader for the statistical voltage monitor of Shunyang brand
Shunyang brand voltage monitor card reader is an optional accessory for Shunyang brand DT1 and DT5 series statistical voltage monitors. When the user uses an electric card or IC card to obtain data and forms a practical file on the computer, a card reader (including a dedicated serial cable, integrated data management system software, electric card or IC card) is required.
The card reader is a signal mode conversion device and cannot be used alone. When using the card reader, you need to use the attached dedicated serial cable to connect the card reader (plugged in from the rear panel) and the computer (need to install the Shunyang brand voltage monitor integrated data management system software) (see Figure 1, Figure) 2) Connect the AC220V power supply, turn on the power switch (the power indicator is on at this time), insert the electrical card or IC card that has obtained the statistical data of the Shunyang brand voltage monitor, and read in on the computer. The process is as follows:
1. Installation of system database and management software
Put the attached Shunyang brand voltage monitor database system and integrated data management system software CD into the computer, follow the instructions and system prompts, first install the database system software/setup.exe, and then install the integrated data management system software version 6.5 /setup.exe. (Note: The operating system must be above win2000, and the computer needs to be restarted after installation. When the system version is upgraded or the system requires reinstallation, you must first use "Add or Remove Programs" to delete the original integrated data management system, and then install a new one Management system.)
2. Initial state setting and maintenance of the integrated data management system
Open the interface of the integrated data management system, and enter the basic content in the "Monitoring Point", "Voltage Level", "User Information", "Subordinate Unit", "Watchmaker" and other fields. (Note that when filling in the content of "monitoring point", "communication method" must be selected as "IC card"). As soon as the situation changes, supplements or amendments must be made in time.
3. Use electricity card or IC card to obtain data from online voltage monitor
As shown in Figure 1, insert the electric card or IC card into the corresponding card socket of the online Shunyang brand voltage monitor (at this time the indicator on the electric card will flash for a few times and then turn to a dim state) to make the voltage The monitor display will automatically go out. After about 1.5 minutes, after the voltage monitor resumes the display (indicating that the reading is completed, the indicator of the electric card is on at this time), then the electric card or IC card can be removed.
4. Use card reader to send data to computer
Turn on the power of the card reader, insert the electric card or IC card that has obtained the information data of the voltage monitor into the corresponding card socket on the front panel of the card reader (see Figure 2, the indicator light on the electric card should be on at this time), Open the integrated data management system, click the big icon "read IC card" in the "system navigation", and click the "read card" button in the dialog box that appears (the card reading indicator on the screen is now flashing, and the card reading is displayed Progress bar); after that, it automatically enters "communication" and "data conversion". When "card reading", "communication" and "data conversion" are over (the three indicator lights turn on and off in turn), it means that the information stored in the card has been Enter it into the computer. At this time, you can insert the next card and click the "read card" button again...
⑴ The electric card or IC card put into use must be numbered, and its number must correspond to the meter number set on the statistical voltage monitor.
⑵ During data access (meter reading), if there is a crash, the card should be removed, and then the voltage monitor protection switch should be turned on, press the "reset" button once when the modification light is off, and wait until the voltage monitor works normally Then re-insert the card to fetch the number. (Warning: If the voltage monitor changes the light and presses the "reset" button when the light is on, all the statistical data in the voltage monitor will be lost.)
6. Correction tips
If there is an error or abnormality when reading the card, please check whether the "communication server" is started, in the "communication server", select whether the COM port of the card reader is connected to the computer is correct, whether the dedicated serial cable is connected correctly, in the integrated data management system Whether the settings are appropriate, whether the computer’s “date and time” and the “year, month, day, hour, minute, and second” in the voltage monitor are consistent with the current date and time, sometimes turning the card reader power off and on will return to normal.